for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||9 February 2016|
|PDF File Size:||15.88 Mb|
|ePub File Size:||13.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 9 of For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
The output frequency is automatically adjusted if you select a function whose maximum frequency is less than that of the currently active function. Inthe amplitude of the carrier is varied by the amplitude of the modulating waveform.
Page and jitter are essentially zero. Joshua, Texas, United States. Application Programs This chapter contains several remote interface application programs to help you develop programs for your own application.
Agilent A Function Generator – Driver – File Exchange – MATLAB Central
Seller information galacticmarketplace You can output the burst at a rate determined by the internal rate generator or the signal level on the rear-panel Ext Trig terminal. This error will be generated in the following instances: Dec 248: Learn About Live Editor. Chapter 3 Features and Functions Amplitude Modulation AM Modulating Waveform Shape The function generator will accept an internal modulating signal, an external modulating signal, or both.
Agileng Delete all user-defined arbitrary waveforms from memory. For example, the following command string guarantees that the first trigger is accepted and executed before the second trigger is recognized. Its combination of mznual and system features makes agilennt function generator a versatile solution for your testing requirements now and in the future. Be the first to write a review. Press or execute TRIG: For example, if you set the stop frequency to 1 MHz in the sine wave function and then change the function to triangle wave, the function generator will adjust the output to kHz the upper limit for triangle waves.
Table Of Contents Contents Chapter 1 Quick Agilebt To prepare the function generator for use 15 If the function generator does not turn on 16 To adjust the carrying handle 18 To set the output frequency 19 To set the output amplitude 20 To set a dc offset msnual 21 To set the duty cycle 22 To output a stored arbitrary waveform Agioent default is 10 Hz.
Page Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code. I had a problem where I was using a power amplifier in conjunction with a A and even with the output off, there was a small leakage that was being amplified.
It does allow you to set the output.
Select a Web Site
Notice that the annunciator flashes, indicating that the displayed Burst parameter is for the burst mode. For this example, you will set the number of cycles for the burst modulation mode.
Features And Functions Features and Functions Mamual will find that this chapter makes it easy to look up all the details about a 3310a feature of the function generator. The function generator is shipped from the factory with a mAT fuse installed. The peak frequency deviation is Hz. Chapter 7 Tutorial Modulation Frequency Sweep The A performs phase-continuous frequency sweeping — stepping from the sweep start frequency to the sweep stop frequency with between 2, and 4, discrete frequency steps.
Learn More – opens in a new window or tab. You can use sine, square, ramp, triangle, or arbitrary waveforms for the hop frequency waveshape.
AGILENT A FUNCTION GENERATOR / ARBITRARY WAVEFORM GENERATOR SERVICE GUIDE | eBay
I could not set the output to 0V to try to mitigate it. Message 8 of Chapter 5 Error Messages Execution Errors Illegal parameter value A discrete parameter was received which was not a valid choice for the command. Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep. Features and Functions You will find that this chapter makes it easy to look up all the details about a particular feature of the function generator.
Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc.
Chapter 7 Tutorial Attributes of AC Signals AC Attributes The magnitude of a sine wave can be described by the value effective heating valuethe peak-to-peak value 2 x zero-to-peakcontinued or the average value. Comments and Ratings 7. The basic principle behind is not unlike an audio compact disc.
Increments the flashing digit. The default is Both internal-external. List the names of all waveforms currently downloaded to memory. Chapter 3 Features and Functions Triggering Single Triggering In the single trigger mode front panel onlyyou can manually trigger the function agilenr by pressing the front-panel key.
Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator.