Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||16 November 2008|
|PDF File Size:||16.79 Mb|
|ePub File Size:||10.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
The oscilloscope should display waveform A. Clean dust from the circuit board with low pressure dry air. The input signal Vi to the RMS Converter is first applied 0840a a circuit which produces a current representative of the absolute value of the input.
Select the KH function and 2 range. Select the range. See Appendix A for Code-to-Name list.
The operation of the ac converter should be checked as follows: Adjust R until the A display indicate one-tenth of the reading observed in step See other items More I used one of my own little 5V adapter. No additional import charges at delivery!
Get the item you ordered or get your money back. When the input required for each figure is applied to the A the manner in which the actual waveform deviates from the expected will indicate the probable cause for the malfunction.
And you have access to the internals. You are covered by the eBay Money Back Guarantee if you receive an item that is not as described in the listing. The number of cycles of oscillator signal that occurred 800a presented on the display as the value of the unknown input being measured by the A. This will be disconnected by default for flukke non rechargeable battery operation.
When making the temperature measurements, observe the following precautions to prevent damage to the probe: The voltage level of the first ms segment of waveforms A2 and B2 should be very close to 1.
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list. Calibrate the Fluke voltmeter.
Fluke 8040A Handheld “Bench” Multimeter Repair/Restore
This procedure can be used as a acceptance test for receiving inspection and as a periodic maintenance -check. The oscilloscope display should be fluje to waveform A2. Select the desired current range in accordance with Table Observe ac current reading in amperes on the A readout. Disconnect the input from the V-H input. This connects the charger to the battery.
Fluke A Multimeter with Manual | eBay
The Model 80F-5 division ratio should be checked every month or every year depending on the fuke requirement. The A display will indicate 0. Contact the seller – opens in a new window or tab and request a shipping method to your location. Select the AC V function and the 2 range. The formula followed to make this calculation is: See terms – opens in a new window or tab. This flukke requires the output of a dc voltage source see Tableset to specified levels, to be applied to the A V-fl and COMMON input while the display is observed to be within specific limits.
Cort vander Lindenstraat Ryswijk Z.
Fluke 8040A Multimeter
Probable cause, U25 defective. The inputs to be measured are connected to the Analog circuitry via the V-H or mA terminals. Where Vi is the value of the ac voltage at any given instant. Select a valid country. If the original container is not available, a new one can be obtained from the John Fluke Mfg. For maintenance for one year or more at an isolated site, it is recommended that at least one of each assembly in the instrument be stocked.
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list. This amount is subject to change until you make payment. A high voltage probe facilitates connection to the measurement source. Flike mV and 2V ranges: To determine the actual surface temperature of a device, both the heat-sinking fljke steady-state errors must be considered. Avoid moving the componentsespecially after the calibration procedure has been completed.
Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc. Estimated on or before Tue. Remove the circuit shield covering the A4 Input pcb and switches for inspection or troubleshooting purposes. Similar sponsored items Feedback on our suggestions – Similar sponsored items. In the resistance K Cl function the reference voltage changes as the input to the A changes.
Repeat steps 41 through 47 until both displays are within specification limits. Operating Instructions All digits are blank.